|Date Added:||1 April 2014|
|File Size:||5.71 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
After above, I opened “ag33xxx – Getting Started 2. You can also troubleshoot and make manual function calls to your instrument to ensure that it is working as expected before moving into automation. Notice that as the mouse is moved over each of the different blocks of the windows, text at labviww top of the window shows what the block contains.
Agilent Technologies / Keysight Technologies 33250A
Select the PC from the SystemDesigner page. So, I’m puzzled about it.
Take note that when the instrument is turned on, the GPIB Address that is assigned to the device is displayed. You can configure the Configure Burst node with several inputs. Download Agilent Series Instrument Driver. The final result will look something like Figures 5 and 6 if done correctly. Hand in this document and the print outs of your choice.
I would very much like to move this code to my current version of Labview which is 7.
Automate Signal Generation From a Keysight/Agilent 33000 Function Generator With LabVIEW NXG
You can then place 3350a one by one on the panel. This way the experimenters can concentrate on the data they are collecting instead of worrying about how everything is set up and changing dials and knobs on all the equipment. Then click the green Run arrow at the top left of the VI.
Hello,I used labview to call a dll, but it have one parameter “pbyFwData” which Pointer to the firmware binary data stream. Thanx for any help.
Windows is a registered trademark of Microsoft Corporation in the United States and other countries. I am attaching the vi. Try it out sometime, because it will come in handy later.
LabVIEW driver for the A | Keysight (formerly Agilent’s Electronic Measurement)
J’ai ajoutdans ma boucle while une autre boucle: If I changed Freq, nothing changed of waveform. The Keithley Multimeter takes bit more work, labvisw the commands are harder. The Front Panel and the Diagram are shown in Figures 1 and 2 respectively. All of the commands that are acceptable for the EA Dual Power Supply are located in the manuals in the file cabinet in the back of the room. Run this vi, I can get correct sine wave from scope.
Would you please send the code to this forum – then we can discuss about it. I have successfully used several Generators NI PXI Waveform Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system.
Most of the commands that will be needed are located in Section 4, Pages Hundreds of built-in functions are included with LabVIEW and the Agilent 33XXX series driver After 3350a the Configure Burst node, be sure to connect the purple instrument wire and the yellow error wire appropriately. If labviea use some clever programming you can get the program to respond just by changing a number here, or the push of a button there.
Waveform Generators | Keysight Community
What if there was a way to set up a computer program to change and measure all of the different measurements? To find out what these numbers mean look in the manual s in Section 4 for the A and the A. Any information that is needed labvied the simulation can be found in the controls and indicators on the Front Panel.
You can also check out the other input and output terminals on the device to see what they are.